A very simple circuit to measure low resistance values from 0.001 up to 1.999 Ohm. With a "Direct Resistance Readout in Ohms". You must use two separate batteries. One for the DMM and one to supply power to the LM317LZ. I recommend the LM317LZ, which is the 100 mA, T0-92 version of the normal LM317. But you can also use the LM317, in the T0-220 package, if you want. The trimpot must be set precisely to deliver 100.0 mA out to get truly accurate resistance measurements.
So you need a very accurate Milli-Amp Meter to adjust this Correctly.
(And like Any Test Equipment, This Calibration should be Re-Checked once a year or so.)
Build Accurate LC Meter and start making your own coils and inductors. This LC Meter allows to measure incredibly small inductances making it perfect tool for making all types of RF coils. LC Meter can measure inductances starting from 10nH - 1000nH, 1uH - 1000uH, 1mH - 100mH and capacitances from 0.1pF up to 900nF. The circuit includes an auto ranging and reset function to make sure the readings are as accurate as possible ... [more]