On the following article learn about Oscilloscope probes, their basic characteristics and proper calibration. Measuring signals with oscilloscopes may be challenging task especially high frequency ones. Without proper oscilloscope probes correct measurement of high speed time domain signals wouldn’t be possible. For high speed measurement you should consider signal parameters like amplitude, source impedance, rise time and bandwidth.
Posted on Sunday, January 30, 2011 • Category: Oscilloscopes
A scope probe is built to minimize ringing by adding resistance. A X1 is better than a piece of co-ax, but a X10 probe is more effective than a X1. A X10 probe has the effect of reducing capacitance by a factor of ten. The trade-off is that is also attenuates the signal by a factor of ten. That is, 1/10 the signal applied to the tip of the probe actually reaches the input of the oscilloscope.
Posted on Thursday, January 27, 2011 • Category: Oscilloscopes
Passive Probe are the most general used scope probe. As the name "passive" suggest, it is made from passive components resistor, capacitor & wires. The leading scope probe maker are LeCroy, Tektronix & Agilent.
Passive probe usually comes with attenuation factor of 1:1, 10:1 and 100:1. Attenuation factor of 1:1 means whatever signal being probe at the probe tip will be shown exactly as it is at the oscilloscope input. So a signal of 1V at the probe tip will be detected as 1V at the scope input.
Attenuation factor of 10:1 means that a signal of 1V at probe tip will be detected as 0.1V at the scope input.
Circuit-Zone.com © 2007-2022. All Rights Reserved.